This Standard specifies the requirements of the component of plasma tubing of the single-use plasmapheresis centrifuge apparatus (hereinafter referred to as tubing), so as to ensure that the centrifugal automatic plasma collection machine is suitable to the plasma separation cup and blood collection apparatus.Wayne Zheng. more than 3s. 5.8.3 Negative blood The filter in pressure monitor connector can block the blood ... 5.9 Flexibility of pump tube The pumping part of tubing shall have the good flexibility. ... when the fixture is closed, it shall block the ventilation of gas and liquid with the pressure of 50 kPa; after opening it, it shallanbsp;...
|Title||:||YY 0326.2-2002: Translated English PDF of Chinese Standard YY0326.2-2002|
|Publisher||:||www.ChineseStandard.net - 2015-05-06|