The charge voltage also tends to pull the drops back to the charging electrode, and reversed flow drops can be generated ... Without the charge- driven dispersion, the interdroplet spacing is often too small, and the problem of coincident particles in the measurement volume occurs. ... (2.5-cm) line, and the nominal drop diameters are 100, 200, and 300 |j.m. Figure 2 is a block diagram of the total ... The first filter is a high-dirt-capacity filter with a nominal rating of 5 M-m - The second filteranbsp;...
|Title||:||Liquid Particle Size Measurement Techniques, 2nd Volume|
|Author||:||E. Dan Hirleman, W. D. Bachalo, Philip G. Felton|
|Publisher||:||ASTM International - 1990|